MLSSA’s technique of Sound Power is derived from post processing of Polar Response of horizontal and vertical axis at 15° interval. Two curves are produced, Sound Power and Directivity Index. After the post processing procedures, the feature “Calculate Level” is used to determine their overall level.
Note: This procedure was design for conventional cone speaker and has been mention by sensored® as not appropriate because the procedure is not suitable for DML’s.
The
name on top of each graphs meaning 


Measurement
System 
: 
MLSSA 
(1)
panel material (3 different kind) 


Mic
to Source Distance 
: 
2
meter 
(2)
panel size in 10cm multiplication 


Acquisition
Length 
: 
3637 
Eg.
sensored
= sensored
or sensored
(mm) 


Antialiasing
Filter 
: 
Chebyshev 
ceiling
= sensored
(mm) 


Bandwidth 
: 
20Hz
to 20kHz 
sensored = sensored (mm) both open back 


FFT
Window 
: 
Rectangular 
all
panel is sensored,
otherwise noted 


FFT
Size 
: 
16384 
all
panels uses sensored
exciters 


Best
Resolution 
: 
3.7Hz 
all
exciters position is sensored
rules 


FFT
Mode 
: 
Sensitivity
8W
 2m 
all
suspension is sensored,
unless stated 


FFT
start to end time 
: 
5.841ms
to 55.839ms 
At
the end of each graph X+Y=Z is estimated level 


Location

: 
Anechoic
Chamber 
The straight line curve is Sound Power, while the dotted line curve is Directivity Index. +Y=13 means scaled up to 20watts RMS, figure on left is Sound Power, figures on right is Directivity Index both calculated by MLSSA.
sensored sensored sensored sensored sensored
79.07+13=92.07 102.82+13=115.82 79.14+13=92.14 104.73+13=117.73
sensored sensored sensored sensored
80.56+13=93.56 103.46+13=116.46 74.82+13=87.82 103.20+13=116.20
sensored sensored sensored sensored
75.25+13=88.25 103.29+13=116.29 74.52+13=87.52 103.36+13=116.36
sensored sensored sensored sensored sensored
80.09+13=93.09 103.96+13=116.96 83.07+13=96.07
On the left is Sound Power by MLSSA, on the right is Power Response by sensored ®. The result was given by sensored sensored in TXT format then imported into MLSSA. The exact same panel is used for both measurements. Both curve looks nearly identical; therefore, the MLSSA method is not entirely doubtable. However sensored ® response is 3dB higher overally. Why? Unknown at the moment, possibly due to summation, averaging difference in method or error here and there, who knows?
sensored sensored sensored sensored sensored sensored sensored sensored sensored sensored
75.97+13=88.97 104.22+13=117.22 78.16+13=91.16 104.75+13=117.75
It is surprising to find the overall level of all panels are not higher than its onaxis response scale up 20watts RMS. For example, sensored sensored have a sensitivity of 90dB, and should achieve 103dB SPL onaxis easily, when “Assuming source is omni directional” it is 103 + 20Log(r) + 10Log(4pr2) = 114dB.
However MLSSA’s technique plus scaling only produce 93.09dB. sensored ®’s Power Response result is also low at 96.07dB. If the summation and average was carry out in Excel, then MLSSA’s Sound Power curve produce 91.81dB, while sensored ®’s Power Response curve produce 94.26dB. Again around 3dB difference. Anyhow the two curve is nearly identical and may suggest sensored ® could be wrong to assume conventional method is not appropriate entirely, given a +3dB off set is needed.
sensored’s method is not entirely different except more elaborate, where it is still based on post processing of horizontal and vertical polar responses but at sensored ° intervals. The elaborate bit is each measurement point is divided by allocated sphere areas describe in AES preprint #4970.
It is unknown at this point how to use the DI curve. Base on the close result of this method and sensored’s, this one is highly adaptable. A question of “Can all results be scaled another 3dB higher to match that of sensored ®’s?” This can be answered in future measurement and analysis.
The sensored’s measurement was based 1V/1m, while the MLSSA’s was singaled at 2.83V/1m. sensored also said MLSSA’s Sound Power conversion is not complete and needs to add . When the suggested correction is applied, the 2 methods yield an exact same level.
ACOUSTICS SECTION AUDIO SECTION CAR SECTION
HOME  Technical Website for Acoustics, Audio and Car